Fluke | 360Back to list
Leakage current clamp meter, 3mA to 60A AC
- Code :
- Brand :
- Manufacturer Ref. :
- Order Ref. :
Display: 3200 pts.
AC current: 3mA to 60A.
Resolution: 1 uA.
Bandwidth: 15Hz - 15kHz.
Jaw opening: 40mm.
|Resolution in nb points :||3,200 counts
|Jaw opening :||40 mm
|Current :||AC (Alternative)
|Min. AC current :||3mA AC (range)
|Max. AC current :||60A AC (range)
|Autonomy :||90 hours
|Safety Level :||CAT I
|Safety Overvoltage :||CAT III 300V
|Package includes :||Soft carrying pouch, Users manual in languages: English, French, Spanish, German, Italian, Portuguese, Simplified Chinese
|Dimension (HxWxD) (mm) :||176x70x25 mm
|Net Weight :||200 g
|Standard Manufacturer Warranty :||1 YEAR
|Bandwidth min-asuppr :||15 Hz
|Bandwidth max-asuppr :||15 kHz
Leakage Current Clamp Meter
The Fluke 360 leakage tester is a rugged and accurate solution for leakage current measurement.
Ideally designed for non-invasive checks
of insulation condition, the Fluke 360 leakage meter allows you to
perform tests without powering down the installation or disconnecting
equipment - saving you time, without compromising on quality or safety.
The unique jaw design of the Fluke 360 AC Leakage Current Clamp Meter eliminates the influence of adjacent current conductors, and minimizes the effects of external magnetic fields, even at low currents. This allows you to perform accurate and reliable tests in today’s crowded electrical environments. The tough and high-quality jaw design of the Fluke 360 leakage tester guarantees high stability for long-term repeatability and durability.
Why leakage current testing?
Leakage currents can be an indicator of the effectiveness of insulation on conductors. High levels of leakage current may be present in circuits where either the insulation resistance is low, or electronic equipment with filters is used. Leakage currents can cause disruptions of the normal operation of equipment and installations. It is possible to locate the source of leakage currents by using the Fluke 360 leakage current clamp meter and taking methodical measurements throughout the installation. The big advantage of leakage current testing versus insulation testing is that the installation and equipment do not need to be powered down but can be done during normal operation.